This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3. This manual provides documentation for the following instruments: PSA Series. EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). Agilent Technologies. PSA Series Spectrum Analyzer. This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3.
|Published (Last):||21 May 2010|
|PDF File Size:||14.10 Mb|
|ePub File Size:||12.99 Mb|
|Price:||Free* [*Free Regsitration Required]|
Used on Option Noise Figure personality. Available only at time of purchase Chapter 5 Description Provides the factory calibration test data and the standard commercial calibration certificate at initial shipment.
In this example you are renaming a trace file; press 3. To access the Copy menu, press boxes as described by the From: Viewing Catalogs and Saving Files Saving a File Saving a File You can save files setups, states, traces, limits, corrections, measurement results, or screens to a floppy disk A: Refer to Figure Using The Front Panel Using the Front Panel Entering Data When setting measurement parameters, there are maual ways to enter or modify the value of the active function: User Preset This type of preset restores the currently selected mode to a known Mode Preset Also, listen for internal fan noise to determine if the analyzer cooling fans are running.
To access the Save menu, press, 2. Page 70 In Case of Difficulty Before calling Agilent Technologies, however or returning an analyzer for servicee444a0 the quick checks listed in the problem. This symbol indicates the input power required is AC.
E4440A PSA Spectrum Analyzer, 3 Hz to 26.5 GHz
This symbol is used to indicate the ON position of the power button. Additional Information Notice The information contained in this document is subject to change without notice. Please be as specific as possible about the problem. This setting enables the alignment routine to run automatically either every 24 hours, or when the internal analyzer temperature changes 3 C. If packing foam is not available, the best alternative is S.
To access the Load menu, press 3. Note that the file extension is always set by the system and for this file type the extension is TRC. This symbol is used to indicate the OFF position of the power button.
Viewing a Signal 1. Make sure there is no disk in the front-panel ailent drive. Enter text from picture: The battery is located on the CPU board.
Agilent Technologies shall not be liable for errors contained herein or for incidental amnual consequential damages in connection with the furnishing, e4440x, or use of this material. Viewing Traces 1, 2, and 3 Step 2. Preselection reduces mixer overload from broadband signals, reduces radiation of local oscillator harmonics back to the device under test, and reduces the level of image and multiple responses displayed.
Copying A File Copying a File 1. Options And Accessories Options and Accessories This chapter lists options and accessories available for your analyzer.
Initial Inspection Inspect the shipping container and the cushioning material for signs of stress. The numeric keypad is also available for the filename.
Directories are displayed as DIR and drives have nothing listed under Type:. Save the file 1. To see that trace 2 is in view mode, blank traces 1 and 3 by pressing.
Retain the shipping materials for future use, as you may wish to ship the analyzer to another location or to Agilent Technologies for service.
The following table describes the four types of spectrum analyzer messages. Front-panel keys that are white perform an immediate action rather than agilentt up a menu.
AGILENT TECHNOLOGIES EA GETTING STARTED MANUAL Pdf Download.
For more information on making measurements, see the measurement guide for your analyzer. This function can be auto-coupled. Making a Basic Measurement This chapter provides information on basic analyzer operation. If the necessary equipment is available, perform the functional tests in the Instrument Messages and Functional Tests document for your analyzer.
Select the type of file you want to rename setup, state, trace, limits, screens, or corrections. Put a formatted floppy in the A: Page 41 Improving Frequency Accuracy Installation And Setup Installation and Setup Protecting Against Electrostatic Discharge Protecting Against Electrostatic Discharge Electrostatic discharge ESD can damage or destroy electronic components the possibility of unseen damage caused by ESD is present whenever components are transported, stored, or used.
Page 75 Table Contacting Agilent Technologies Online assistance: Page 73 Perform an Alignment. Item Name Access features used with all analyzer modes and affect the state of the entire spectrum System Keys analyzer.
Reset the analyzer by pressing 2. Page 30 Front and Rear Panel Features Key Overview In other key menus, one key label will always be highlighted to show which key has been selected but the menu is immediately exited when a selection is made. Check to see if the green LED above the power switch is on. Activate trace 2 and put it into minimum hold by pressing analyzer display should look similar to Figure